Introduction: serie 1 vs serie 2
Serie 1 EHT block cannot be repaired. Serie 2 EHT block can be heated to remove wax. Here is what a serie 1 EHT block looks like. Note it is installed with connections on the left side, not on top:
Refurbishing EHT block
Look at section “part list” to order necessary parts.
Then make your PCB, as illustrated below (reference: “SDS Labs” @ quadesl.com, page 66 here: http://quadesl.com/origRefs/quad_book.pdf):
Output for bass panel may seem a bit low but not loaded with the panels themselves. Therefore, less current can go to the HV probe. The multimeter is in ‘hold max’ position. That said, shown legacy design exhibits slightly higher output on bass panel: wax has positive impact ? higher value for capacitors ? leakage in diodes ? Nevertheless, no need to worry: from reference site quadesl.org, 4.9kV is a very good value, and this increases once panels are connected. Besides, 3 instances of this re-building measure exactly the same.
A more compact version is probably possible using 1 higher grade diode like GP02-40 instead of 2 standard 1N4007 diodes. Another option is to use same PCB for either version: 16 1N4007 or 8 GP02-40, but you lose benefit of compact size. Here is a proposal here below that features:
- compact size: 8 diodes only
- higher capacitance option: 8 additional capacitors can be added on the other side of the board
231 total views, 2 views today